[Download not found]
Notification chains are being configured with too many message receivers and too short of a repeat duration, which causes encoder queue logjams, caregiver burnout, and damages facility’s trust in the product.
TECH SUPPORT SUMMARY:
Below is a comparison of the disparity between two (Non-hypothetical) scenarios that accomplish the same thing. Each has 10 CNA’s and an escalation at 3 minutes that adds 5 nurses for a total of 6 minutes of alerts. Scenario one uses common capcodes and a repeat duration of 90 seconds. Scenario two uses individual capcodes and a repeat duration of 60 seconds. As each page requires approximately 4 seconds of processing, if the incident is not restored for the entire 6 minutes it’s the difference between 4 pages that takes a total of 12 seconds of processing and 75 pages that takes up the entire 6 minutes. In scenario two, if you were to receive a second alarm during this 6 minute window it is highly probable that it will fail to send them on time rendering the notification configuration inoperable.
- Discuss specific notification requirements/expectations with facility prior to installation.
- Trim down your notifications by using common cap codes.
- You should never have more than 3-4 message receivers per link.
- Your “Seconds between repeat notifications” should typically be no less than 120 seconds.